Oxford Dipper Manual Pool hoist with Ranger Seat
- 3 Floor Fixing Options
- Use with almost any swimming pool
- Safe Working Load (SWL) of 140 kg (22 st)
- Spreader Bar and Sling option
- Ranger Transporter option
Safe Working Load (SWL): 140 kg / 22 st
Boom Length: 1498 mm
Height (socket A & B, boom lowered): 1422 mm
Height (socket A & B, boom full elevation): 2108 mm
Base of Seat Above Floor Level (socket A & B): 889 mm
Base of Seat Below Floor Level (socket A & B): 914 mm
Height (socket C, boom lowered): 1572 mm
Height (socket C, boom full elevation): 2258 mm
Base of Seat Above Floor Level (socket C): 1039 mm
Base of Seat Above Floor Level (socket C): 764 mm
Weight - Mast Assembly: 15.4 kg
Weight - Boom Assembly: 10.4 kg
Weight - Chair Support Tube: 3.6 kg
Weight - Ranger Seat: 4.5 kg
Weight - Ranger Chassis: 8.7 kg
Weight - Spreader Bar Assembly: 1.7 kg
There are three types of base sockets for the dipper to be located into either Socket A, B or C enabling the Dipper to be fitted alongside the vast majority of swimming pools, Hydrotherapy pools or boat mooring pontoons.
The base socket of the Dipper is subjected to extremely high loads when in use. It is essential the immediate area around the socket is solid concrete construction, we recommend 1 meter sq. We strongly suggest if you’re unsure of what’s required, please contact us on 01772 814555.
All installations should be tested against the manufacturers recommendations prior to hand over to the users, with particular attention being paid to the socket and surrounding concrete.
1. Dipper Socket A
This is the most commonly supplied socket and is fitted into a hole 90mm diameter, core bored into the concrete slab to a depth of 245mm. The socket is held firmly in place with a non-shrinking cement. When correctly fitted, the top of the socket is 12mm below the floor level. Each socket is supplied with a blanking plug to prevent an ingress of water when the mast has been removed for storage and injury to swimmers or other participants. When fitted, the top of the blanking plug sits flush with the floor level.
2. Dipper Socket B
This socket is for use on swimming pools with a water channel around the perimeter of the pool. The “B” socket is an “A” socket with a plate which adds additional bracing. The “B” socket plate is fixed in place with six 15mm diameter expanding bolts of the rawl plug type.
3. Dipper Socket C
This socket is surface mounted and stands proud of the floor by 150mm. It is used mainly where there is insufficient depth of concrete for “A” or “B” sockets. The “C” socket plate is drilled for four 15mm rawl bolts. The recommended minimum depth of concrete for the “C” socket is 150mm.
The “C” socket adds 150mm to the height of the mast. This alters the maximum height of the bottom of the chair (to 1039mm) and the lowest position of the chair bottom below floor level (to 764mm).
AAL LTD are accredited suppliers, installers and service providers for the Oxford Dipper Pool Hoist. We provide a complete range of lifting & handling products and services to both domestic and trade customers.
Stephen Pritchard, Managing Director of AAL LTD is a full-time wheelchair user and is driven to supply a quality service at an affordable price. The Oxford Dipper Pool hoist is the most predominant hoist within the UK’s NHS hospitals, Care Centres, Private Residential Homes, Yachting & Sailing Clubs, Quaysides, Therapy & Swimming Pools and Health Centres which has a maximum SWL (Safe Working Load) of 22 Stone (140Kg).
The Dipper uses a hydraulic pump to raise and lower service users and gives the support workers and service users peace of mind as it’s not reliant on electricity. To operate the hoist, simply pump the hydraulic pump using the handle to raise the user and to release, unscrew the black pump valve to slowly lower the service user.
There are two variations of the Oxford Dipper Pool Hoist, the standard Oxford Dipper with spreader bar and the Oxford Dipper with Ranger seat. Both variations are available with an array of sockets to meet each individual customers specification.
Oxford Dipper Pool hoist with Ranger seat
The Ranger seat version is the most popular as it offers dignity to those service users who need extra assistance in a changing or bedroom before entering the pool. An example of this would be if a hotel had a pool with Oxford Dipper Pool hoist and a guest wished to use the pool, they could then get dressed in their room and then be pushed to the pool side and transferred into the pool safely and without fuss.
The Oxford Dipper Ranger has been cleverly designed to aid users who need assistance approaching and entering the pool. The chair is reversed up to the Dipper pool hoist and the ‘T’ bar is connected to the back of Ranger chair. The support person simply pumps the dipper until it locks into position and continues to pump, raising the service user up further, enabling the seat to detach from the Ranger chassis and leave the service user sitting in the seat section.
When lowered into position in the pool, the service user can then float away from the Ranger seat if desired.
The process is carried out in reverse to lift the person from the pool, very easy, safe and convenient for both the user and support person whilst maintaining dignity.
Can you have both options on one Dipper?
There are two types of Oxford Dipper boom extensions, the 6 point spreader bar and the Ski tube which is compatible with the Oxford Ranger Seat. The 6 point spreader bar is used with slings, slings with loops, not clips. The spreader bar is very easy to attached, it takes 2 minutes.
The Ski Tube connects again, very easy. This enables the Ranger seat to reverse up to lift, connect and lift with ease within a few minutes. To demonstrate how easy it is to connect these attachments, we've created a couple of short videos for your convenience.
How do you connect the Oxford Dipper Spreader Bar?
How do you connect the Oxford Dipper Ski Tube for the Ranger seat?
How is the Dipper fitted to the floor?
The Dipper comes with three (3) options. Socket A, B and C. The sockets are ergonomically designed to be fit for different materials and finishes enabling the Oxford Dipper to be installed in almost any pool side setting. For more information on the socket fittings, please visit our website here.
Socket A for Oxford Dipper
To fit the socket A, a 1M sq cube of concrete is required as the minimum recommendation. If already installed, a 91mm hole then needs to be drilled 350mm deep.
An accredited engineer will then install the sockets interior, casing and socket cap which will then sit flush with the floor.
The Dipper then slots into the socket when in use and once the use has been completed or you wish to store away, simply slide the dipper out of the hole ensuring you have lowered the boom to its lowest point and with the dipper strap, secure the pump tightly to the mast. This will prevent damage to the mast, pump and release valve when transporting the dipper to storage.
After use please ensure the dipper pump is lowered to its lowest possible position, this ensures that the chrome on the hydraulic pump remains oiled and rust free and therefore extending the service life of the hoist. Should the hoist not remain oiled, the chrome will become pitted and this will destroy the internal seals over a long period of time, a new hydraulic dipper pump costs approx. £500.00 exclusive of vat.
Socket B for Oxford Dipper
Socket B is fitted and installed as shown in the picture above, socket B is primarily used in swimming pools which have a water channel around that outlines the perimeter of the pool. Socket B is a replica of socket A with an added base plate which adds additional bracing and support to the Oxford Dipper.
The B socket plate is fixed into place with six 15mm diameter expanding bolts of the rawplug type, for more information please see the socket construction as illustrated above.
As with the A socket, the B
socket is slightly below floor level and is fitted with a blanking plug. The
blanking plug can be inserted into the hole when the Oxford Dipper is
transported away for storage, this leaves a clean flush finish to the pool
sides surface, minimising injury and maximising hygiene in the pool area.
Socket C for Oxford Dipper
Socket C is surface mounted and stands proud of the floor by approx. 150mm, this socket is used mainly where there is insufficient depth of concrete for sockets A or B.
The C socket plate is drilled with four 15mm Rawplug type expanding bolts, the recommended minimum depth of concrete to enable an accredited engineer to install the hoist is 150mm.
Due to the pretruding nature of the socket, socket C adds 150mm to the overall height of the hoist and therefore alters the maximum height of the bottom of the chair to 1039mm and the lowest position of the chair bottom below floor level to 764mm. For more information, please see the illustration found above.
Servicing, Parts and Maintenance
AAL LTD offers a suite of service and maintenance packages to suit your ue and needs, the following description below is a brief guideline of our services:
A 3.1 service contract plan offers you an annual service inclusive of the required LOLER test andcertificate and the appropriate weight testing for three (3) consecutive years.
A 3.2 service contact plan offers you a six (6) monthly inspection, followed by a six (6) monthly service inclusive of the required LOLER test and certificate and the appropriate weight testing for three (3) consecutive years.
As we are an accredited Oxford UK Dealer, we are able to offer you the complete range of spare parts for all Oxford products here. Should you require any advice or assistance then please do not hesitate in calling us on 01772 814555, we are here to help!